Crompton Potentiometer complete setup for physics experiment. The slide-wire type of potentiometer described in the previous article is not a practical form of construction. The long slide wire is awkward, and. Potentiometers. Introduction. A Basic dc Potentiometer. Crompton’s dc Potentiometers. Applications of dc Potentiometers. AC Potentiometers.
|Published (Last):||20 April 2012|
|PDF File Size:||10.88 Mb|
|ePub File Size:||16.21 Mb|
|Price:||Free* [*Free Regsitration Required]|
The value of unknown emf is read off directly from the settings of the dial adjust slide wire. Single range potentiometer using independent calibrating circuit. Amarnath Reddy I am a graduate in electrical engineering and blogger by passion.
Find more suppliers in Faridabad Carbon Potentiometer in Faridabad. The main dial setting is kept nearly equal to the emf being measured. The resistances of the middle dial shunt two of the coils of the first dial. Direct reading type, improved from with standardizing coil consisting of 8 coils and a circular slide wire The scale is divided into parts with potential drop of mill volts Fine and coarse rheostats are provided for adjusting the slide wire current Range 0.
This potentiometer uses the Kelvin Varley arrangement shown below. This unit has been specially designed for hospitals and physiotherapy centers where the machine is required to run constantly for many hours to survive with the rush of patients.
The figure below shows a deflectional potentiometer.
dw crompton potentiometer pdf converter
The limitations imposed on the performance of ordinary potentiometers by slide wire are eliminated in a vernier potentiometer. Thus a considerable change in standard cell emf owing to temperature changes can be allowed for the operation of the circuit below.
This unit has been specially designed for hospitals and physiotherapy centers where the machine is required to run constantly for After completion of standardisation, the switch ‘S’ is thrown to operate potsntiometer thereby connecting the unknown emf into the potentiometer circuit. Supposing the value of emf of the standard cell is 1.
Get Best Price Request a quote. Modern laboratory type potentiometers use calibrated dial resistors and a small circular wire of one or more turns, thereby reducing the size of the instrument. The vernier potentiometer reads to the increment of 0. Year of Establishment The following steps are used when making measurements with the above potentiometer: However, in this potentiometerthere are three measuring dials.
The long slide wire is awkward, and even for the length shown cannot be read to a very great degree of precision. The dial settings are kept exactly the same as in the original standardisation process.
However, for some applications higher precision and accuracy are needed, requiring changes in design features.
When the range switch S is thrown to position x 0. Sw this potentiometer, only one or two main dials, consisting of decade resistance boxes, are used. These difficulties to get accurate results mainly arise because of lack of uniformity of slide wire and the difficulties in maintaining the contacts in good shape.
Some of the features are explained below: The standardising circuit may be checked for the constancy of working current, anytime during the measurements by simply throwing the switch S back to calibrate position. This condition is satisfied only when the total battery current has the same value for each measuring range.
The slide wire is provided with scale divisions and since the total resistance of slide wire corresponds to a voltage drop of 0.
Call Send a quick message. Final adjustments are made for zero deflection with the help of the rheostat. The operation of the duo range potentiometer potentiomteer be more easily understood and analysed by redrawing in its simplified form by omitting the galvanometer and calibration standardising circuit.
The measuring circuit current I crlmpton must be equal to 10 mA in order to produce a voltage drop of 1. This fixes the working current potentipmeter its proper value. The circuit of a single potentiometer may be modified in a simple way to add a second range, which is usually by a second factor, such as 0. Rust proof body Durability Easy to operate Specification: View Contact Call Seller Now. Digital Potentiometer in Ambala. These EMFs can be minimised only by special construction i. This potentiometer is provided with a double throw switch which allows the connection of either the standard cell or the unknown emf to be applied to the working circuit.
The selector switch S is put in the calibrate position and the rheostat is adjusted so that there is no current flowing through the galvanometer. The protective resistance is kept in the circuit in the initial stages so as to protect the galvanometer from being getting damaged. A centre zero type galvanometer is included in the circuit to potentiometef deflection. The slide-wire type of potentiometer described in the previous article is not a practical form of construction.
All modern potentiometers incorporate a separate standard cell dial circuit calibrating circuit. The figure below shows a separate standard cell dial incorporated in a single range potentiometer. The output of the thermocouple is fed to the unknown emf terminals.